CZY-GY Primary Adhesive Tester (ChP) is designed according to adhesion test method of Chinese Pharmacopoeia and applicable to tack test of medical plasters and medical patches, etc.
In the determination of tack by rolling-ball method, a steel ball is released at the top of an inclined surface, then allowed to accelerate down and roll on an adhesive tape on the inclined surface. Tack is determined by the maximum size of the steel ball which can be stopped by the adhesion of the adhesive specimen.
Chinese Pharmacopoeia 2015 (Adhesion Test Method), GB/T 4825
|Applicable to tack test of medical plasters and patches, etc.|
|Test Angle||0 ~ 45°|
|Panel Width||120 mm|
|Test Area Width||80 mm|
|Standard Steel Ball||1/32” ~ 2” (46 Steel Balls)|
|Instrument Dimension||580 mm (L)×150 mm (W)×180 mm (H)|
|Net Weight||6 kg|
Note 1: The parameters in the table are measured by professional operator in Labthink laboratory according to relative requirements for laboratory standard conditions.
Note 2: The described product features, test standards and applications should be in line with Technical Specifications.